Sahyog Wellness Portable Traveler Mesh Nebulizer with USB Port
12 x 4.3 cm, CE Approved, convenient, noiseless, easy to operate and carry device for effective respiratory relief Product ID: 3338912 -
No Return Accepted for this product
- Power supply - To operate with alkaline batteries, open the battery cover and insert 2 alkaline aa batteries. Close the cover and use. You can also use USB cable provided with the box, kindly connect the USB cable to the adaptor or power bank to use this device. Pouring medicine into the box - open the medicine chamber, pour in the medicine (max 8 ml, min 0.5 ml quantity) and tighten the cover. Keep the medicine liquid and mesh plate in constant contact to prevent mesh plate air shock
- How to use the nebulizer - Push the start button. Indicator light will turn blue. If the nebulizer is switched on without introducing medicine liquid, indicator will show yellow light and nebulizer shall automatically turn off. Indicator light turning red shows low voltage, thus changes with new alkaline batteries. If using mouthpiece, take the mouthpiece flat end inside the mouth cavity. Inhale from mouth and exhale from nose cavity to direct medicine mist to targeted area
- How to use the nebulizer - If using face mask, hold the mask close to face and inhale deeply from nasal cavity and exhale from nasal passage. Tilt the nebulizer to ensure constant contact of medicine liquid with the vibrating mesh plate. When the medicine is exhausted , the nebulizer will turn off automatically. After using the nebulizer, press the start/stop button to turn off the device. De-assemble the mouthpiece / mask.
- How to use the nebulizer - Press the push button behind main unit and slide out the medicine bottle. Empty the medicine bottle of any residue and immediately proceed for cleaning and disinfecting. Cleaning medicine chamber - open the medicine bottle and discard any residual medicine left. Pour a small amount of distilled water between 30 to 60 *c. Long press the start/stop button for 5 seconds. Yellow blue and red light, indicating cleaning mode is on
- Cleaning medicine chamber- Run the device in cleaning mode for 1 to 2 mins, nebulizing distilled water to remove medicine from mesh plate holes. Rinse the medicine bottle & cap with distilled water. Wipe off excess water with a clean cloth and air dry. Disinfecting the bottle can be done at the end of each day of usage. Immerse the medicine bottle in 75% medical alcohol for 3 mins to disinfect & then air dry. Cleaning main unit- wipe the main unit with 75% medical alcohol solution
Brand - Sahyog Wellness
Type - Portable Nebulizer with USB Port
Product dimensions - 12 x 4.3 cm
Battery - 2 alkaline AA batteries
Items included in the Package:
Nebuizer device, packet, product manual, user guide, adult mask, child mask, USB cable
Country of Origin: China
The product expiry date will be displayed only when the pincode is entered.
Questions & Answers
Your Question has been sucessfully submitted.
You will be notify by an email once as soon as answers your questions.
OTHER OFFERINGS FROM Sahyog Wellness
Tell us if we missed any relevant information on the products?
Thank you for your Feedback
Shop with Confidence
Hassle Free Returns